ii Heracell Service Manual – Free download as PDF File .pdf), Text File . txt) or read online for free. incubator heracell SM. HERAcell® CO2 Incubator Operating Instructions. Version: / – 3 -. Contents. 1. Information regarding Safe Operation. Manuals and User Guides for Thermo Scientific Heracell i. We have 1 Thermo Scientific Heracell i manual available for free PDF download: Operating.
|Published (Last):||6 December 2013|
|PDF File Size:||3.80 Mb|
|ePub File Size:||17.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Door Switch The outer door can only be closed after the glass door has been latched properly.
Thermo Scientific Heracell i Manuals
See detailed information in calibration. Page 25 The height of the floor stand should not fall below mm.
I have one of these CO2 inbators that has condensation building up around seal on the inside of the glass door. For thermal separation, an adapter plate  can be placed between the two devices.
Page User menu structure: The control loop error monitoring system optional is enabled.
Examples Of Data Logger Codes 1. By looping the gas supply through, up to 4 devices can be supplied by a CO gas supply. Notice The contra-con decontamination routine must be performed every 3 24i.
Supply interface Supply interface Figure. Sensor System O sensor are integral to the control system of the device.
Thermo Scientific Heracell 240i Operating Instructions Manual
The connection of the device to an external alarm system must only be carried out by adequately trained Instruction Of The Operating Personnel Heracel receptacle for the two device fuses is integral to hearcell socket. The pipe of the pressure compensation opening must not be extended or redirected.
Wipe out the rests of the water at the floor plate of the water reservoir. EXE to open this file. Starting the installation routine: Page Glass door seal, measurement opening Tempered silicone Sensor block Stainless 24i0 1. Transport After transport or storing under humid conditions a drying-out process must be performed. After transport or storing under humid conditions a drying-out process must be performed. No tools are required to replace the seal.
Thermo Scientific – i and i CO2 Community, Manuals and Specifications | MedWrench
Gas Connection Set the gas supply to a range between 0. To exit the error display: The cross sections of the openings are smaller when samples are accessed.
Do not use damaged components when connecting the device to the power supply! CO gas cylinder 8. Key for opening the menu navigation 6. Calibration should be performed with the CO set value designated for the work process prospective work process value. Page Failure to set the gas-tight screen function in accordance with the actually installed door version may impair the incubation performance.
Distributor connection for looping through the O gas supply for up to 3 additional devices independent of device type Installing The Shelf System Tools are not required for the installation of the shelf system. Gas tight screen Figure. Page 51 The support rails  of the shelf system are perforated every 42 mm so that the support hooks  can be inserted variably for any culture container size required.
The center strut is stabilized by the spring pressure. The switching state of the O nanual loop does 2240i affect the function of the optional gas monitoring system.